Tandberg 3000MXP Tactical - inproper use caused damage
Looking at those pictures it's hard to see what damage is there, other than one of the cables has become unattached. You may have some luck in re-soldering the yellow cable back on to the connector, which as Martin has said above, should be able to be done at your local electronics repair shop (or, if you're confortable with doing it yourself, give it a go).
Given that it's just low voltage for the IR, you could try holding it in place and testing the IR to see it works or not before proceeding with the solderting.
Please remember to rate responses and to mark your question as answered if appropriate.
Wayne -- Please remember to rate responses and to mark your question as answered if appropriate.
These are the paths to get to each CCX logs through CLI. They may be helpful if you are having issues accessing RTMT or downloading logs through it.
If you want to download them you have to prefix "file get " and you can add one of the options (re...